After installing and configuring your device correctly, if your still seeing "No Data Series" under the status tab there might still be an issue. This guide will cover some of the most common causes of this error.
The most common cause of this is simply not giving the software enough time to generate a reading. The default interval is set at 600 seconds (10 minutes). This means you need to give the software 10 minutes before a reading is captured and displayed. You can change this interval under the settings tab. Please remember to click "Save & Restart" after making changes, and allow the software some extra time after restarting.
The main thing to look for in a successful install is that your device is detected on COMX in the software. You should also be able to see the device under Device Manager on your PC. If the device is detected, the software is working properly and just needs time to actually take a reading.
If your device is not detected, please follow this guide:Finding The Latest Drivers
After installing the software and/or drivers, some computers will require a reboot in order to run the device properly. This is especially true if upgrading from older versions of either. If none of the previous stepped have solved your issue, a simple reboot of your PC should get things up and running.
Readings in the software are disabled when using Insights/Sensor Cloud mode. Version 4.5 is in Insights/Sensor Cloud mode by default. Click "Settings" then click "Switch to Developer Mode". After you click "Save and Start" a reading should be visible after the set interval(default is 600 seconds/10 minutes).